ASTM G_Laboratory Immersion Corrosion Testing of Metals – Free download as PDF File .pdf), Text File .txt) or read online for free. ASTM G Other ASTM procedures for laboratory corrosion tests are tabulated in the Appendix. NOTE 1—Warning: In many cases the corrosion product on the reac-. Find the most up-to-date version of ASTM G31 at Engineering
|Published (Last):||22 April 2005|
|PDF File Size:||1.34 Mb|
|ePub File Size:||6.10 Mb|
|Price:||Free* [*Free Regsitration Required]|
It should be used as a reference to ensure that the test will allow generation of data relevant to g13 application with the minimum of interferences. Warning- In many cases the corrosion product on the reactive metals titanium and zirconium is a hard and tightly bonded oxide that defies removal by chemical or ordinary mechanical means.
Link to Active This link will always route to the current Active version of the standard. These factors include specimen preparation, apparatus, test conditions, methods of cleaning specimens, evaluation of results, and calculation and reporting of corrosion rates. Active view current version of standard. The values given in parentheses are for information only.
It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. This guide also emphasizes the importance of recording all pertinent data and provides a checklist for reporting test data.
Link to Active This link will always route to the current Active version of the standard. The values given in parentheses are for information only. These factors include apparatus, sampling, test specimen, test conditions test solution composition, temperature, gas sparging, fluid motion, solution volume, method of supporting test specimens, duration of testmethods of cleaning test specimens, interpretation of results, and calculation of corrosion rates.
ASTM G31 – 72() Standard Practice for Laboratory Immersion Corrosion Testing of Metals
This practice also emphasizes the importance of recording all pertinent data and provides a checklist for reporting test data. Corrosion testing by its very nature precludes complete standardization.
It is impractical to propose an inflexible standard laboratory corrosion testing procedure for general use, except for material qualification tests where standardization is required.
Referenced Documents purchase separately The documents listed below are referenced within the subject standard but are not provided as part of the standard.
Historical Version s – view previous versions of standard. This standard does not purport to address all of the safety concerns, if any, associated with its use.
One purpose for this guide is to promote better correlation of results in the future and the reduction of conflicting reports through a more detailed recording of meaningful factors and conditions.
In many such cases, corrosion rates are established by mass gain rather than mass loss. This standard, rather than a standardized procedure, is presented as a guide so that some of the pitfalls of such testing may be avoided. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior aatm use.